Setup is in the AP Sets-you can modify those as you see fit. See attached. General rule of thumb that most use is FMax 80x rpm and 1600 or 3200 LOR (today's computers can handle the data load). I would advise against 800 lines as if you have a large FMax you get poor resolution that will make it difficult to determine specific frequencies during analysis. Collect better data and you will be rewarded. I would suggest you also routinely correct a PeakVue reading on each rolling element bearing also.