[Added subject matter] Two proximity probes mounted at 90° on a relatively small shaft can exhibit cross talk. Since the probe tip spacing varies based on probe type, it is best to check the probe OEM manual for recommendations.
Dear Experts, A question just clicked my mind that can there be a disturbance in eddy current probe output, if a magnet mount sensor is placed near it during portable data collection(Casing Vibrations)? Regards
There would be minimal effect on a proximity probe from a sensor magnet in close proximity. For one thing it would be a static non-varying magnetic field. If we further look at the case of a varying magnetic field, thus a field able to create a voltage given the presence of a conductor in that field, then the chart below shows the effect of a 300 Gauss 60 Hz field on a proximity probe mounted on a generator. It shows that for 5-meter proximitor, probe and cable at a 50 mil gap, the total...
It appears that the SKF CMSS 65 Eddy probes that we have used for years are no longer available! The new equivalent are CMSS 75 probes, but does anyone know if I could use a 75 probe, with the rest of a 65 system (Driver and Extension Cable)?
When you block a person, they can no longer invite you to a private message or post to your profile wall. Replies and comments they make will be collapsed/hidden by default. Finally, you'll never receive email notifications about content they create or likes they designate for your content.
Note: if you proceed, you will no longer be following .