We can group several measurement by setting it in the dbase setup. Go to the dbase setup tree structure, highlight on the equipment, then select edit. Select Group/Ch tab on the right side of the screen. There will be list of your measurement point, for your example pick a number let's say 21 for group number on M1H and M1P and for channel column also 1 (because you group 2 measurements for 1 sensor i.e. the hor position and peakvue, if you have want to take 2 channel measurements for example M1H & M1V simultaneously using 2 sensors, then put 1 and 2 respectively on the ch column.)
Or just hit the help button on the AMS MHM Software, it always helpful
When you block a person, they can no longer invite you to a private message or post to your profile wall. Replies and comments they make will be collapsed/hidden by default. Finally, you'll never receive email notifications about content they create or likes they designate for your content.
Note: if you proceed, you will no longer be following .