Hi, I cannot understand this message on CSI -2140?? this message comes to screen several times before data acquisition when i press the enter after mounting sensor. even i have created a good database according to equipment's description e.g rpm. kw amp load etc in AMS Suit
You cannot have a peakvue measurement where your Fmax (upper frequency in the Spectrum parameter) is higher than the High pass filter (HP filter setting in signal processing parm).
You can see it up in database setup in the AP sets. it usually happens when the Upper frequency is in orders. You can either lower it or put in in Hertz/CPM.
Beatnik? how to change settings, can you please help me in details??
Why don't you hit the "Yes" button as suggested on the screen shot, then go in and look at the setup to see what has changed. Then ask yourself "why" so you learn from the experience.
Rizwan Sulehri posted:I changed analysis parameters settings but when ever i want to take gearbox readings this message still appears? why?
I'm not real familiar with the 2140 so just a guess that when doing a gearbox, you input the tooth counts, which then set up filters to include mesh frequency.
As John stated if you configure the machine with all of the pertinent parameters the software will do the work for you. If that is not the case calculate estimated highest GMF of interest manually and setup the point accordingly to eliminate the annoying error messages. There are quite a few PeakVue instruction papers out on the web that give detailed instructions on point setups.
It is telling you that your high pass filter you have selected in your database is too low, and gives you the option of selecting yes and the analyzer will automatically select next higher filter. As above posters have said, you need to take a look at your highest fault frequencies ex: BPFI capture of 4 harmonics or Gear mesh frequency and however many harmonics of that you wish to capture.For example, if your FMAX is higher than your filter selected then you will see this message. Say your FMAX is 50,000 CPM and you have a high pass filter selected of 500HZ then it will give you this message telling you that you can either push enter and it will go to the 1000HZ filter, or you can change the AP setting. I went into my database and altered the AP settings to the correct ones so I would no longer see this after going from 2130 to 2140, some of my coworkers just push enter and go on.